DENTAILAIRE - BEFORE AND AFTER
Basic Maintenance • Machine and Handpiece Disassembly • Replacement (H2O) Filters • Replace Air Filter • Replace Oil Filters • Replace Intake Filter • Replace/Drain Oil • Cleaning and Lubrication of Handpieces • Recalibration of Handpiece Output • Cleaning of Control Panel • Regulator Calibration • Maintenance Procedures
IM3 - BEFORE AND AFTER
Basic Maintenance • Machine and Handpiece Disassembly • Replacement (H2O) Filters • Replace Air Filter • Replacement (HEPA) Filter • Replace Intake Filter • Cleaning and Lubrication of Handpieces • Cleaning of Control Panel • Regulator Calibration • Calibration of Handpieces • Replacement of IM3 toggle switch
*NEW* IM3 MOTOR INSTALL
Customer complaint of dental IM3 machine becoming audibly 'louder' and taking significantly longer to build pressure. After diagnosis, a new motor was needed along with some significant cleaning and a maintenance kit. See the progression from deconstruction to complete install and gauge holding pressure. Annual maintenance can be performed to keep your machine running for years to come.
*OLD* Henry Schein (VetBase)
Complete customization • Customer wanted to 'cannibalize' two old dental units to create one 'like new unit.' • JunAir Compressor was gutted and cleaned with new oil and components • Old 'control panel' was swapped with a new panel including: Water Bottle Mount, In-Line water port, scaler integration, and new handpieces • This unit was a complete overhaul but is now serviceable and ready for years of use • Old equipment can compete with modern counterparts.